|Year : 2017 | Volume
| Issue : 6 | Page : 505-507
Spontaneous arterial catheter fracture and embolisation: Unpredicted complication
Suresh Babu Kale, Senthilkumar Ramalingam
Department of Cardiothoracic Surgery, Meenakshi Hospital, Thanjavur, Tamil Nadu, India
|Date of Web Publication||12-Jun-2017|
Suresh Babu Kale
Department of Cardiothoracic Surgery, Meenakshi Hospital, Tirchy Main Road, Thanjavur - 613 005, Tamil Nadu
Source of Support: None, Conflict of Interest: None
Arterial cannulation is a common procedure in the care and management of critically ill patients. Blood pressure measurement, arterial blood sampling and cardiac output determinations are a few primary reasons for arterial cannulation. This invasive catheter placement also imposes certain risks and clinical management problems like bleeding, thrombosis and hematoma formation. Fracture and embolisation of arterial catheters placed in the femoral region is a rare and serious complication that may result in arterial occlusion and thrombosis. Percutaneous retrieval of the foreign body is frequently reported to be the technique of choice, leaving the surgical option to a small group. We report two consecutive cases of catheter fracture that was successfully retrieved by a combination of surgery and balloon tip retrieval device.
Keywords: Arterial-catheter, complications, embolisation, fracture
|How to cite this article:|
Kale SB, Ramalingam S. Spontaneous arterial catheter fracture and embolisation: Unpredicted complication. Indian J Anaesth 2017;61:505-7
|How to cite this URL:|
Kale SB, Ramalingam S. Spontaneous arterial catheter fracture and embolisation: Unpredicted complication. Indian J Anaesth [serial online] 2017 [cited 2019 Oct 16];61:505-7. Available from: http://www.ijaweb.org/text.asp?2017/61/6/505/207758
| Introduction|| |
Arterial cannulation is commonly practiced while managing critically ill patients. Blood pressure measurement, arterial blood sampling, cardiac output determinations and a site for rapid blood withdrawal are the primary reasons for arterial cannulation. In the era of rapidly advancing technology, there is an increase in the number of catheter-based procedures and therefore the incidence of catheter based complications. One of the potential complications is the fracture and dislodgement of a femoral artery catheter used for routine perioperative monitoring. Catheters used in cardiac theaters are sheath-less, small in size and are removed prior to ambulating the patient and hence catheter fracture and embolisation, though a well-known complication for central venous catheters, is rarely seen. Catheter quality and manufacturing defects though rare are seen frequently in current cost saving era when catheters are subjected to unconventional use.
| Case Reports|| |
A 54 year old male patient with critical two vessel coronary artery disease and poor ventricular function underwent successful bypass surgery with right common femoral artery and left radial artery cannulation for blood pressure measurement and possible placement of intra-aortic balloon pump. A single lumen Unicath™ dialysis catheter [Newtech Medical Devices, New Delhi, India] was used via the femoral route to measure the arterial pressure. The patient was shifted to the Intensive Care Unit and extubated the next morning. Bleeding was noticed in the right groin and on examination the femoral catheter was found fractured at its junction with the hub [Figure 1]. The proximal segment of the catheter was not palpable. The distal pedal pulses were normally felt. The catheter had embolised and it was not clear in what direction. After control of the common, superficial and profunda vessels, the patient was heparinised and common femoral artery opened along the puncture site. The re-exploration of the right groin was done at the bedside with sterile technique. A number 4F Fogarty catheter [Edward Lifesciences LLC, CA, USA] was passed along the distal arterial route and on multiple passages, the embolised segment was retrieved. The common femoral artery was repaired and the right groin closed in layers. The patient was discharged with good right femoral and right pedal pulse.
A 65 year old male with triple vessel coronary artery disease and borderline ventricular function was induced for bypass surgery and the right femoral artery catheter was cannulated for arterial pressure measurement. Whilst positioning the patient for painting and draping, the femoral catheter fractured at its junction with the hub with resultant bleeding from the fracture site. The proximal catheter had migrated distally and required a groin cut down to the level of common femoral artery for retrieval of the fractured end and repair of the artery. The fractured end was still in the puncture site and could be retrieved with an artery forceps. The artery was repaired and the groin wound suture closed.
| Discussion|| |
Arterial cannulation is a common procedure in the care and management of critically ill patients and most commonly placed in the radial and femoral arteries. Invasive catheter placement can impose certain risks and clinical management problems. Femoral artery is cannulated by various catheters varying in texture, length and caliber with or without guidewire assistance. In 1967, Mortensen  reported complications from 3193 arterial needle punctures, cannulations and incisions. The overall complication rate was 13%, including 66 major and 321 minor complications. Catheter fracture was evidently not reported in this series.
Catheter fracture with subsequent embolisation is rare and sometimes a late complication of central venous catheter placement, reported to occur in 0.5% to 3% of patients, either spontaneously and at times with port removal., It is commonly reported with subclavian approach and postulated to be preceded by the pinch-off sign. The complication could not be eliminated with internal jugular approach as well., The fracture was attributed to repeated compression of the catheter against the clavicle and regular wear and tear. Spontaneous catheter fractures are underestimated as some vascular access devices are not assessed after completion of scheduled treatment and may never be identified as damaged. Majority are asymptomatic. Although there are various signs reported to be classical for a central venous catheter fracture,,, no such signs are reported for a femoral approach. Ideally catheter material needs to be sufficiently stiff to maintain structural integrity and flexible to minimise discomfort or injury. Teflon and polyethylene are the two most common materials used in arterial catheters. A bent or kinked catheter is a common observation and may be a precursor for fracture. Catheter length has not been studied as a factor related to fracture and one article recommended a 3.2 to 5.1 centimeter catheter for radial and brachial artery cannulation and a 16 centimeter catheter for femoral artery cannulation. Literature review of complications of femoral arterial cannulation found no incidence of catheter fracture, making it a rare complication. A defective catheter escaping quality control checks in the production line might also be a reason for it to fracture under mechanical strain. Relevant to our assumption is a report by Panuccio et al. of 5 episodes in 4 patients of epidemic spontaneous catheter rupture. They suspected a problem with defective silicone catheters during the manufacturing process because no definite cause for the fractures could be found and most ruptures occurred at similar sites shortly after initiation of peritoneal dialysis. Alternatively, damage to the catheter at the time of insertion, such as an accidental needle-stick injury or use of a hemostat to clamp the catheter, might also contribute to an internal break. A thorough inspection and careful manipulation of the catheter before implantation is therefore required to avoid this unusual complication and the resultant patient morbidity. In our case the new catheter transection occurred spontaneously at the junction of the catheter with its hub which is its weakest part. We have since then discontinued the use of dialysis catheters for routine femoral artery cannulations and switched over to small guide-wire based shorter catheters and have not found any untoward incident.
Percutaneous retrieval of intravascular foreign bodies is a safe technique that is effective, accessible and standardised. Many retrieval devices have been designed , and their success rate has varied from 50% to nearly 100%. Fluoroscopic identification of fractured catheter can be difficult and ultrasound examination of the vascular tree may help locate the embolised segment and assist in planning surgical course. Surgical retrieval is the best intervention  in cases where catheter fragment lacks a free end, is entrapped in deep peripheral branches, is endothelialised or has escaped outside the vessel walls. We have successfully used a combination of both surgery and retrieval balloons to extract the embolised segment, making it an innovative technique.
| Conclusion|| |
Low-maintenance systems must be researched and developed to avoid the complications of indwelling arterial lines. Simple and reproducible techniques are to be practised to avoid serious complications in critically ill patients.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Wu CY, Fu JY, Feng PH, Kao TC, Yu SY, Li HJ, et al.
Catheter fracture of intravenous ports and its management. World J Surg 2011;35:2403-10.
Mortensen JD. Clinical sequelae from arterial needle puncture, cannulation, and incision. Circulation 1967;35:1118-23.
Lin CH, Wu HS, Chan DC, Hsieh CB, Huang MH, Yu JC. The mechanisms of failure of totally implantable central venous access system: Analysis of 73 cases with fracture of catheter. Eur J Surg Oncol 2010;36:100-3.
Aitken DR, Minton JP. The pinch-off sign: A warning of impending problems with permanent subclavian catheters. Am J Surg 1984;148:633-6.
Jensen MO. Anatomical basis of central venous catheter fracture. Clin Anat 2008;21:106-10.
Shimizu T, Mekata E, Murata S, Yamamoto T, Tani T. A case of catheter fracture of a totally implantable access port introduced through the right internal jugular vein. J Surg Oncol 2011;103:460-1.
Mirza B, Vanek VW, Kupensky DT. Pinch-off syndrome: Case report and collective review of the literature. Am Surg 2004;70:635-44.
Panuccio V, Enia G, Crucitti S, Pustorino M, Biondo A, Zoccali C. Another 'epidemic' of spontaneous rupture of peritoneal catheters. Nephron 1998;79:359.
Wu JR, Hsu JH, Chang TT, Dai ZK, Lu CC, Wu DK. Nonsurgical percutaneous retrieval of dislodged Port-A catheters from pulmonary artery in children. Jpn Heart J 2002;43:295-300.
Liu JC, Tseng HS, Chen CY, Chern MS, Chang CY. Percutaneous retrieval of 20 centrally dislodged Port-A catheter fragments. Clin Imaging 2004;28:223-9.
Cheng CC, Tsai TN, Yang CC, Han CL. Percutaneous retrieval of dislodged totally implantable central venous access system in 92 cases: Experience in a single hospital. Eur J Radiol 2009;69:346-50.
Chang TM, Pellegrini D, Ostrovsky A, Marrangoni AG. Surgical management of entrapped percutaneous transluminal coronary angioplasty hardware. Tex Heart Inst J 2002;29:329-32.
Dondelinger RF, Lepoutre B, Kurdziel JC. Percutaneous vascular foreign body retrieval: Experience of an 11-year period. Eur J Radiol 1991;12:4-10.